1 Implementation of the 2012 ACR CT QC Manual in a Community Hospital Setting BRUCE E. HASSELQUIST, PH.D., DABR, DABSNM ASPIRUS WAUSAU HOSPITAL
2 Conflict of Interest Disclaimer Employee of Aspirus Wausau Hospital ACR Activities Reviewer for CT accreditation program (physics) Reviewer for NM accreditation program (physics) Member Medical Physics Guidelines Committee No conflicts to disclose
3 Objectives Review the major elements of the 2012 ACR QC manual Compare the manual to current ACR accreditation program and phantom testing requirements and to the current technical standard Highlight some interesting details of the new physicist testing recommendations
4 2012 ACR CT Quality Control Manual Table of Contents Radiologist s Section Definitions of QA and QC Responsibilities Radiologic Technologist s Section Daily QC Weekly QC Monthly QC Medical Physicist s Section Annual QC
5 To be accredited by the ACR Effective 12/01/2013, all facilities applying for accreditation must maintain a documented QC program and must comply with the minimum frequencies of testing outlined in the 2012 ACR Computed Tomography Quality Control Manual. Page 4: 2012 ACR CT QC Manual
6 What s Mandatory? Should s (177 instances!) Acceptance testing before 1 st patient scanned Equipment evaluation dated within 1 year at renewal Evaluation after major repairs, incl x-ray tube and detector assembly replacement Refer to state and local regulations if more restrictive Greater frequently for QC tests when being introduced or results inconsistent. Records accessible from CT scanner location Must s (35 instances!) Maintain documented QC program Comply with minimum testing frequencies QMP oversees QC program QMP monitors performance at installation and annually Annual equipment evaluation submitted initially and at renewal Evaluation dated within 14 months at renewal Evaluation performed by a QMP Annual QMP review of tech QC records (qtly preferred)
7 Quality Assurance is a comprehensive concept to ensure that: 1. Every imaging procedure is necessary and appropriate to the clinical problem at hand; 2. The combination of acquisition parameters used for each exam is appropriate to address the clinical question; 3. The images generated contain information critical to the solution of that problem; 4. The recorded information is correctly interpreted and made available in a timely fashion to the patient s physician; and 5. The examination results in the lowest possible risk to the patient and is consistent with Objective 2 (above).
8 Quality Control is a series of distinct technical procedures Four steps are involved: 1. Acceptance testing to detect defects in equipment that is newly installed 2. Establishment of baseline equipment performance 3. Detection and diagnosis of changes in equipment performance before they become apparent in images 4. Verification that the causes of deterioration in equipment performance have been corrected
9 (Supervising) Radiologist Radiologists must assume the primary responsibility for the quality of CT and for the implementation of an effective QA program at their site. Convene a CT QA or Protocol Review Committee Oversees QA program Sets goals and directions Determines policies Assesses effectiveness of QA activities Assist with development of CT QA Manual Participate in Interpretive Quality Assurance aka peer review
10 Quality Assurance Committee Design and review all new or modified CT protocol settings to ensure that both image quality and radiation dose are appropriate Develop internal radiation dose thresholds during any new CT protocol design. Implement steps to ensure patient safety and to reduce future risk if an estimated dose value is above the applicable threshold for any routine clinical exam Review, at least annually, all protocols to ensure no unintended changes have been applied that may degrade image quality or unreasonably increase dose Establish a policy stating that the CT dose estimate interface option is not to be disabled and that the dose information is displayed during the exam prescription phase
11 Good Practices Established by the Team 1. Technologists are provided access to adequate training and continuing education in CT that includes a focus on patient safety. 2. An orientation program has been provided for technologists based on a carefully established procedures manual. 3. A technologist has been selected as the primary QC technologist to perform the prescribed QC tests. 4. Appropriate test equipment and materials necessary for the technologist to perform the QC tests have been provided 5. Staffing and scheduling are arranged so that adequate time is available to carry out the QC tests and record and interpret the results.
12 Good Practices Established by the Team 6. A qualified medical physicist will review the technologist s test results at least every 3 months or more frequently if consistency has not yet been achieved. 7. A qualified individual has been designated to oversee the safety program for employees, patients, and other individuals in the surrounding area. 8. Established protocols will be followed 9. Corrective action procedures will be followed when images of poor quality are presented for interpretation
13 Good Practices Established by the Team 10. Radiologists will participate in the Radiology Department s practice improvement program 11. Documentation of current qualifications will be provided by all interpreting radiologists in accordance with ACR Accreditation and local rules.
14 CT QA Procedures Manual 1. Who is responsible for QC? 2. What are your QC procedures? 3. Where are the records of all QC tests, service reports, etc? 4. How are your CT operators trained? 5. How is your scanner maintained? 6. What CT protocols are in place? Contrast? Positioning? 7. Do you have minutes from your CT QA committee? 8. What are your policies for pregnant patients and staff? 9. What are your procedures for disinfecting equipment?
15 Technologist QC Effective 12/01/2013, all facilities applying for accreditation must maintain a documented QC program and must comply with the minimum frequencies of testing outlined in the 2012 ACR Computed Tomography Quality Control Manual. Daily Water CT# and standard deviation Artifact evaluation Weekly Wet laser printer QC Monthly Visual checklist Dry laser QC Display monitor QC (See also current ACR CT accreditation requirements document.)
16 Daily QC The QMP should assist with setting up QC protocols and determining pass / fail criteria. Should acquire a set of standard artifactfree images for comparison with daily QC. Must use daily QC scan protocols to acquire these images. Test Water CT Number and Standard Deviation (noise) Artifact Evaluation Procedure Axial and Helical scans ROI in center of image at center of scan as well as at leading or trailing edge of fan beam CT# Criteria Water: 0 ± 5 HU( must ) ACR Phan: 0 ± 7 HU( must ) Axial scan Thinnest available axial images across the full z-axis extent; 2 (LS16) or 3 (VCT) scans, if necessary Recommend periodic use of larger uniform phantom (manufacturer s or 32 cm dosimetry phantom)
17 Weekly QC To be performed if film is used for primary interpretation. Test Wet Laser Printer QC Procedure Film SMPTE pattern 6 on 1 in all 6 frames. Monitor OD s in upper left image for 0%, 10%, 40% & 90% gray level patches. Evaluate film for artifacts. Should be done for each sending modality, e.g. CT and MRI. Target OD s and control limits are provided in manual.
18 Monthly QC Laser printer QC to be performed if film is used for primary interpretation. Backup printers may be tested prior to clinical use as well as when initiating QC program. Display monitors should be recalibrated at least annually. All QC records should be reviewed and signed by QMP as part of equipment evaluation. Retain records for 5 yr. Test Visual Checklist Dry Laser Printer QC (same as for Wet Laser Printers) Display Monitor QC Procedure See QC manual for recommended list of items to check. Film SMPTE pattern 6 on 1 in all 6 frames. Monitor OD s in upper left image for 0%, 10%, 40% & 90% gray level patches. Evaluate film for artifacts. Should be done for each sending modality, e.g. CT and MRI. Target OD s and control limits are provided in manual. Display SMPTE (or TG18-QC) on acquisition monitor. Evaluate 5% & 95% patches and other gray level patches.
19 Automatic QC Procedures Automatic analysis SW may be used but must be approved by QMP. Automatic QC procedures may be used in place of these tests if the Qualified Medical Physicist has critically reviewed them and approved this substitution (in writing). It is not recommended that an automatic QC process be considered as a replacement for the artifact analysis portion of daily QC.
20 Physicist (QMP) Our role Establish continuous QC program for facility Review QC program results annually Equipment evaluations at acceptance and annually Communicating test results and recommending corrective action are areas that can be improved in the practices of most QMPs. Required tests Missing tests Relegate to acceptance?
21 Annual Physics QC Effective 12/01/2013, all facilities applying for accreditation must maintain a documented QC program and must comply with the minimum frequencies of testing outlined in the 2012 ACR Computed Tomography Quality Control Manual. Review of clinical protocols Scout / alignment light accuracy Image thickness Table travel accuracy Radiation beam width Low-contrast performance Spatial resolution CT number accuracy Artifact evaluation CT number uniformity Dosimetry Display monitor performance
22 Review of Clinical Protocols Recommendations provided in QC manual. Team should design and review all new or modified protocol settings Institute a regular review process of all protocols At least 6 protocols (ped / adult: head & abd; high res chest; brain perfusion, if used) Review appropriate use of dose reduction methods Review clinical scans for: Image quality Dose Centering
23 Scout / Alignment Light Accuracy 1. Align phantom with laser and do axial scan. 2. Scout phantom and scan at marker. Scan location relative to alignment light or scout prescription should be accurate to within 2 mm. ACR CTAP Phantom Module 1 or similar (see AAPM Rpt #39)
24 Image Thickness Perform axial scans of the phantom for each reconstructed image thickness used clinically. Image thickness should be within 1.5 mm of nominal thickness. Good practice mandates helical scans (SSP s) at acceptance mm 2.5 mm 10 mm ACR CTAP Phantom Module 1 or similar 5 mm 1.25 mm.
25 Table Travel Accuracy Check known distance and accuracy of return to initial position after travel to max distance. Translation accuracy should be within 2 mm. ACR CTAP Phantom or any phantom with 2 sets of external fiducial markers of known separation (see AAPM Rpt #39)
26 Radiation Beam Width Scan strip at isocenter using each unique NxT product (beam collimation) available. Measure FWHM of beam profile. Compare to manufacturer s standards or ACR criteria of within the greater of 3 mm or 30% of the nominal beam collimation. Note: 120 kvp, 200 mas works well with GAF Chromic film.
27 Low-Contrast Performance Establish correlation to ACR phantom for alternate phantoms. Equivalent to ACR accreditation LCR test protocol, analysis and criteria. Additionally, 6 mm targets must be visualized for adult head and abdomen. Corrective action is immediate. Scan Protocol Adult Head 1.0 Pediatric Head 1.0 Adult Abdomen 1.0 Pediatric Abdomen 0.5 CNR (ACR) ACR CTAP Phantom Module 2 or any phantom with low contrast objects of known contrast.
28 Spatial Resolution Evaluate HCR for relevant clinical exams, e.g. adult abdomen and highresolution chest. Corrective action is immediate. Scan Protocol Adult Abdomen High-Resolution Chest Limiting Resolution 6 lp/cm 8 lp/cm ACR CTAP Phantom Module 4 or any phantom with high contrast objects of known resolution.
29 80 kvp 100 kvp CT Number Accuracy Scan phantom using all 4 ACR clinical protocols. Evaluate CT # in at least three target materials for all 4 scans. 120 kvp 140 kvp Scan water (or water equivalent) phantom at all clinical kvp s. Use ACR accreditation criteria for ACR phantom. ACR CTAP Phantom Module 1 or similar phantom with water, air and 1 other material
30 2.5 mm 16i (40 mm beam collimation) W/L: 20/0 Artifact Evaluation Same test as for daily technologist QC. Perform axial scan(s) using the thinnest available axial slices, spanning the z-axis extent of the detector array. For a GE VCT this could include: mm, 16i (10 mm) 1.25 mm, 16i (20 mm) 2.5 mm, 16i (40 mm) Water phantom provided by CT manufacturer or ACR CTAP Phantom Module 3.
31 CT Number Uniformity ACR recommends using technologist artifact QC protocol. Analysis and criteria are the same as for the ACR accreditation program. Water phantom provided by CT manufacturer or ACR CTAP Phantom Module 3.
32 Dosimetry In head holder for adult head; On table for pediatric head and abdomen protocols. Equivalent to ACR accreditation dosimetry measurement protocol, analysis and criteria. Additionally, compare measured CTDI vol to the values reported by the scanner. Agreement should be within 20%. Be sure to note if scanner display value based on phantom other than the one use for measurement. Year-to-year variation should be no more than 5%. On table for adult abdomen protocol. 16 cm (head) and 32 cm (body) CTDI dosimetry phantoms
34 Display Monitor Performance GSDF should be visually consistent year to year. (see also TG18 report) Observe: 1) 5%/95% patches 2) Line pair patterns 3) Black-white transitions 4) Loss of bit depth 5) Geometric distortion Measure: 1) Min and max brightness (0%, 100%) 2) Response curve (GSDF) 3) Center / edge brightness uniformity Min (0%) 1.2 cd/m 2 Max (100%) 90 cd/m 2 Luminance nonuniformity should be no more than 15% for FPDs SMPTE pattern or equivalent (e.g. TG18 QC)
35 Survey Report Form
36 Automatic QC Software Automatic analysis SW may be used but must be verified. CT Scanner Acceptance Testing Perform all tests independent of manufacturer s QC SW Must compare results to SW if it is to be used during annual testing Must verify SW if it was not evaluated during acceptance testing Annual Performance Evaluation Manufacturer s automatic QC SW may be used if previously verified by QMP
37 Additional Physics Tests These tests are not mentioned in the QC manual but could be done at acceptance. Some manufacturer s provide specifications and CTDI adjustment factors that apply to these measurements. Comparison can also be made to console CTDI display values. Image Noise CTDI Accuracy Output Reproducibility Output kvp dependence mr/mas Linearity Output Beam Thickness Dependence EKG ma Modulation Slice Sensitivity Profile Beam Angulation Accuracy Scatter
38 Slice Sensitivity Profile SSP measurements are recommended as part of acceptance testing in the slice thickness section of the ACR QC manual. Collimation (mm) Slice Thickness (mm) Recon Interval (mm) Max in ROI Slice Sensitivity Profile 1.25 mm slice Distance (mm) Meas FWHM (mm) Manuf. Spec. (mm) (Plus mode) (Plus mode) ACR CTAP Phantom Module 3 or any phantom with a small high contrast object (0.28 mm for ACR) in a uniform background.
39 2012 ACR-AAPM CT Medical Physics Technical Standard 2012 ACR CT QC Manual Limited protocol review Image localization from SPR Alignment light accuracy Table incrementation accuracy Reconstructed image thickness Radiation beam width Image quality HCR, LCR, Uniformity, Noise, Artifact CT number accuracy Dosimetry (CTDI vol, patient dose) Acquisition workstation display Safety evaluation, scatter at AT Review of clinical protocols Scout/alignment light accuracy Table travel accuracy Image thickness Radiation beam width Spatial resolution Low-contrast performance CT number uniformity Artifact evaluation CT number accuracy Dosimetry Display monitor performance
40 Minnesota Rules ACR CT QC Manual Accuracy of scout localization view Table backlash & Table indexing Accuracy of distance measurements CT # uniformity and artifacts CT # calibration and noise CT dose index CT # dependence on slice thickness Hard copy output & visual display Review of clinical protocols Scout/alignment light accuracy Table travel accuracy Image thickness Radiation beam width Spatial resolution Low-contrast performance CT number uniformity Artifact evaluation CT number accuracy Dosimetry Display monitor performance
41 How are we doing? Modified annual equipment evaluation protocols Convened CT QA committee Modified our technologist QC program and associated forms not implemented at WI sites Revised policies & procedures at MI sites Created draft site specific CT QA manual Implementing ACR NRDR/DIR program in Wausau Installing and implementing Dose Check SW Continuing to review protocols (esp. pediatric) QC phantom images analyzed monthly by QMP
CT Quality Control Manual FAQs General Question: How often will the QC Manual be updated and how will those updates be communicated? Answer: The ACR CT Physics Subcommittee will review any comments, issues
BICOE Stereotactic Breast Biopsy and Breast Ultrasound Accreditation William Geiser, MS DABR Senior Medical Physicist MD Anderson Cancer Center Houston, Texas email@example.com 1 Introduction Objectives
Computed tomography Acceptance testing and dose measurements Jonas Andersson Medical Physicist, Ph.D. Department of Radiation Sciences University Hospital of Norrland, Umeå Sweden Contents The Computed
BICOE Stereotactic Breast Biopsy and Breast Ultrasound Accreditation William Geiser, MS DABR Senior Medical Physicist MD Anderson Cancer Center Houston, Texas firstname.lastname@example.org 1 Introduction Objectives
ESTABLISHING DRLs in PEDIATRIC CT Keith Strauss, MSc, FAAPM, FACR Cincinnati Children s Hospital University of Cincinnati College of Medicine CT Dose Indices CTDI INTRODUCTION CTDI 100, CTDI w, CTDI vol
Computed Tomography-Basic Principles and Good Practice X-Ray & CT Physics / Clinical CT INSTRUCTORS: Dane Franklin, MBA, RT (R) (CT) Office hours will be Tuesdays from 5pm to 6pm CLASSROOM: TIME: REQUIRED
MQSA - Who s Who Update on MQSA and Mammography Accreditation The Law: Mammography Quality Standards Act (MQSA) The Regulator: US Food and Drug Administration (FDA) Priscilla F. Butler, M.S. Senior Director,
American College of Radiology CT Accreditation Program Testing Instructions (Revised January 6, 2017) This guide provides all of the instructions necessary for clinical tests, phantom tests and general
ACR Accreditation Update in Mammography Eric Berns, PhD University of Colorado Hospital Denver Health Medical Center Denver, CO *No financial disclosures to report Overview New ACR Activities Requirements
MRI FAQs Application - General Q. My facility plans to apply for ACR MRI Accreditation, where do I start? A. Start by reading the following documents, available on the ACR website: The Diagnostic Modality
The American College of Radiology Breast Ultrasound Accreditation Program: Frequently Asked Questions (Revised: July 31, 2017) Table of Contents APPLICATION - GENERAL... 1 MOVED FACILITIES AND UNITS...
ACR Accreditation Update in MRI Whole Body Systems Extremity (MSK) Ron Price Vanderbilt University Medical Center Nashville, TN Dedicated Breast MRI Accreditation Update 1. ACR MRI Accreditation: Overview,
Nuts and Bolts of MRI Physics Tests or Selling Preparation H Robert A. Bell, PhD (858) 759-0150 email@example.com Robert A. Bell, PhD Since 1987 RAB has provided independent consulting services to
Implementation of a New Tomosynthesis Program: A Physicists Perspective Bill Geiser, MS DABR Senior Medical Physicist firstname.lastname@example.org 1 Conflict of Interest None of the authors nor their immediate
CT Head Dose Reduction Using Spiral Scanning Protocol Reed, William J MD; Broderick, Daniel F, MD; Weindling, Steven M, MD; Czervionke, Leo F MD; and Morin, Richard L; PhD; Mayo Clinic, Department of Radiology;
Managing Pediatric CT Patient Doses Keith J. Strauss, MSc, FAAPM, FACR President X-Ray Computations, Inc. Boston, Massachusetts Acknowledgements Marilyn Goske, MD John Boone, PhD Cynthia McCollough, PhD
Varian Edge Experience Jinkoo Kim, Ph.D Henry Ford Health System Disclosures I participate in research funded by Varian Medical Systems. Outline of Presentation Review advanced imaging in Varian Edge Linear
ORAL SURGERY GROUP P.A. PRACTICE LIMITED TO ORAL AND MAXILLOFACIAL SURGERY DENTAL IMPLANTS, WISDOM TEETH, ORAL PATHOLOGY, AND ORAL MEDICINE Dr. David M. Rauch Dr. Philip S. Engel Dr. Richard K. Stern Dr.
MVCT Image Guidance and QA Robert Staton, PhD DABR MD Anderson Cancer Center Orlando ACMP Annual Meeting 2011 Disclosures MDACCO has received grant funding from TomoTherapy, Inc. Overview TomoTherapy MVCT
Current Ultrasound Quality Control Recommendations and Techniques Evan J. Boote, Ph.D. University of Missouri-Columbia Thank you for your interest - today I hope to provide some basic recommendations for
Patient Centering for MDCT CT Imaging Original Research Jianhai Li 1 Unni K. Udayasankar 1 Thomas L. Toth 2 John Seamans 2 William C. Small 1 Mannudeep K. Kalra 1,3 Li J, Udayasankar UK, Toth TL, Seamans
What is radiation? Radiation is energy emitted from a source, that travels through space and can penetrate matter. Listed below are two types that we are exposed to and contribute to our overall radiation
Ask EuroSafe Imaging Tips & Tricks CT Working Group Optimization of scan length to reduce CT radiation dose Alban Gervaise (Centre Hospitalier Universitaire Nancy, FR) Mika Kortesniemi (HUS Medical Imaging
Preparing for Medical Physics Components of the ABR Core Examination The ABR core examination for radiologists contains material on medical physics. This content is based on the medical physics that is
Cardiac CT - Coronary Calcium Basics Workshop II (Basic) J. Jeffrey Carr, MD, MSCE Dept. of Radiology & Public Health Sciences Wake Forest University School of Medicine Winston-Salem, NC USA No significant
Dental Intraoral X-ray Systems PROPOSED REVISIONS TO 4732.XXXX, 2.0 4732.#### DENTAL INTRAORAL X-RAY SYSTEMS; STATIONARY AND MOBILE. Commented [JC(1]: Based on part 4732.0880. Subpart 1. Applicability.
HISTORY OF MQSA AND ACR DEBORAH THAMES R.T. (R)(M)(QM) WHY MQSA? In the United States, there was a lack of standards in mammography imaging. Reporting Imaging Type of imaging screening/diagnostic Equipment
To Shield or Not to Shield? Lincoln L. Berland, M.D. Disclosures Consultant to: Nuance, Inc. Page 2 Breast Radiation on CT Use of chest CT has increased in women vulnerable to cancer induction by radiation.
Three Dimensional BANG Polymer Gel Dosimeters Marek J. Maryanski MGS Research, Inc. Guilford, CT Educational objectives: Describe the need for high-resolution 3D dosimetry in 3D CRT. Explain the physics
45 HR PET/CT REGISTRY REVIEW COURSE Course Control Document Timothy K. Marshel, MBA, R.T. (R), (N)(CT)(MR)(NCT)(PET)(CNMT) The PET/CT Training Institute, Inc. SNMMI-TS 028600-028632 45hr CEH s Voice Credits
Overview of Quality Assurance in Proton Therapy Omar Zeidan AAPM 2012 Charlotte, NC July 30 st, 2012 Learning Objectives Understand proton beam dosimetry characteristics and compare them to photon beams
The New Jersey Radiographic Quality Assurance Program at 5 Years Julie Timins, MD a, Paul Orlando, BS b, Jill Lipoti, PhD b Purpose: Five years ago, the New Jersey Bureau of Radiological Health decided
Patient dose assessment of CT perfusion scanning at the RSCH Lesley Leavesley, Emma Whitehead, Matthew Pryor, Debbie Peet Regional Radiation Protection Service Royal Surrey County Hospital, Guildford Overview
Acknowledgments Thanks to Saiful Huq for an illuminating conversation about the application of TG-100 Jennifer O Daniel, Ph.D. & Fang-Fang Yin, Ph.D. Duke University Medical Center Annual AAPM Meeting,
ULTRASOUND QA SOLUTIONS Ensure Accurate Screening, Diagnosis and Monitoring DOPPLER FLOW PHANTOMS MULTI-PURPOSE PHANTOMS TRAINING PHANTOMS INNOVATORS IN ADVANCED ULTRASOUND TECHNIQUES Gammex is the only
Mammography What is Mammography? Mammography is a specific type of imaging that uses a low-dose x-ray system to examine breasts. A mammography exam, called a mammogram, is used to aid in the early detection
Dual-Energy Imaging of Bone Edema on a Dedicated Multi-Source Cone-Beam CT System for the Extremities W Zbijewski, 1 A Sisniega, 1 JW Stayman, 1 N Packard, 2 J Yorkston, 2 G Thawait, 3 S Demehri, 3 J Fritz,
ecommons@aku Imaging & Diagnostic Radiology, East Africa Medical College, East Africa January 2012 Establishing the quality management baseline in the use of computed tomography machines in Kenya Geoffrey
Scan for mobile link. Mammography Mammography is a specific type of breast imaging that uses low-dose x-rays to detect cancer early before women experience symptoms when it is most treatable. Tell your
CT Dose Reduction in Pediatric Patients By Kelly Firestine, RT(R)(CT)(M) Executive Summary CT is an incredibly valuable imaging tool, but there are unique concerns with pediatric patients, including the
Managing the imaging dose during Image-guided Radiotherapy Martin J Murphy PhD Department of Radiation Oncology Virginia Commonwealth University Radiographic image guidance has emerged as the new paradigm
Thoracic examinations with 16, 64, 128 and 256 slices CT: comparison of exposure doses measured with an anthropomorphic phantom and TLD dosimeters Poster No.: C-2584 Congress: ECR 2015 Type: Scientific
DOELAP On-Site Assessment Requirements Checklist Page 1 of 24 General Requirements Y, N, N/A Demonstration of Conformance G.1 Latest version of protocols or procedures G.2 Latest version of dosimeter specifications
The American College of Radiology Breast MRI Accreditation Program: Frequently Asked Questions (Updated: October 4, 2017) To navigate in this document, either click a section in the Table of Contents or
Don t Electrocute Me!: Common Misconceptions in Imaging and Radiation Safety (and What to Do About Them) Rebecca Milman Marsh, Ph.D. University of Colorado Department of Radiology Who in the Facility Works
The Bone Densitometry Examination The purpose of The American Registry of Radiologic Technologist (ARRT ) Bone Densitometry Examination is to assess the knowledge and cognitive skills underlying the intelligent
Scan for mobile link. X-ray (Radiography) - Chest Chest x-ray uses a very small dose of ionizing radiation to produce pictures of the inside of the chest. It is used to evaluate the lungs, heart and chest
Scan for mobile link. Intravenous Pyelogram (IVP) Intravenous pyelogram (IVP) is an x-ray exam that uses an injection of contrast material to evaluate your kidneys, ureters and bladder and help diagnose
Patient Dose Estimates from CT Examinations Patient Dose Estimates from CT Examinations John M. Boone, Ph.D. Professor and Vice Chairman of Radiology Professor of Biomedical Engineering University of California,
The IAC Standards and Guidelines for Dental CT Accreditation Table of Contents All entries in Table of Contents are linked to the corresponding sections. Introduction... 3 Part A: Organization... 4 Section
Computed Tomography The purpose of structured education is to provide the opportunity for individuals to develop mastery of discipline-specific knowledge that, when coupled with selected clinical experiences,
Code of Practice for Radiation Protection in Dentistry Code of Practice For Radiation Protection in Dentistry 10 OCTOBER 2017 CONTENTS 1. INTRODUCTION... 3 1.0 CITATION... 3 1.1 BACKGROUND... 3 1.2 PURPOSE
Quality Assurance in Radiation Therapy Purpose (from George) To make aware of the needs for quality assurance in radiation therapy To identify components of a quality assurance program in radiation therapy
Automated CT Protocol Design Advantages and Pitfalls of Algorithm-Based Technique Selection in Pediatrics Robert MacDougall, M.Sc. Department of Radiology Boston Children s Hospital Disclosures 2 Learning
Scan for mobile link. Brachytherapy Brachytherapy places radioactive sources inside the patient on a temporary or permanent basis to damage cancer cells DNA and destroy their ability to divide and grow.
RADIATION SAFETY Junior Radiology Course Expectations for the Junior Radiology Course Medical School wants students to learn basic principles, factual knowledge, safety info, etc. Medical Students want
CONTENT SPECIFICATIONS The Computed Tomography Examination The purpose of The American Registry of Radiologic Technologists (ARRT ) Computed Tomography Examination is to assess the knowledge and cognitive
Highmark Privileging Application INSTRUCTIONS 1. Complete this application if you provide any diagnostic imaging services. 2. Please complete a separate application for each physical location where imaging
Digital Mammography Imaging on KODAK CR Systems Digital mammography imaging from Carestream Health solutions for great workflow, productivity, and patient care. Commercial distribution of the CR Mammography
Tracking Doses in the Pediatric Population Frederic H. Fahey DSc Boston Children s Hospital Harvard Medical School email@example.com Disclosures Sadly, none that pay me any money! SNMMI
Patient / Organ Dose in CT Patient specific and organ dose estimation H.D. Nagel Dr. HD Nagel, Science & Technology for Radiology Buchholz / Germany www.sascrad.com 1 Topics CTDI & patient dose SSDE Organ
Measurements of Air Kerma Index in Computed Tomography: A comparison among methodologies Thêssa C. Alonso 1, 2, Arnaldo P. Mourão 1, 3, Teógenes A. Da Silva 1, 2 1 Program of Nuclear Science and Techniques
Kent State University Catalog 2017-2018 1 RADIOLOGIC AND IMAGING SCIENCE (RIS) RIS 34001 INTRODUCTION TO DIAGNOSTIC MEDICAL SONOGRAPHY 1 Credit Provides an introduction to diagnostic medical sonography.
PHYS 383: Applications of physics in medicine (offered at the University of Waterloo from Jan 2015) Course Description: This course is an introduction to physics in medicine and is intended to introduce
MRI to fit your planning Philips Panorama HFO Oncology Configuration MR Imaging that fits Philips Panorama HFO Oncology Configuration allows radiation oncologists to take full advantage of MRI s excellent
Implementing New Technologies for Stereotactic Radiosurgery and Stereotactic Body Radiation Therapy Implementation of radiosurgery and SBRT requires a fundamentally sound approach Errors don t blur out
Bone Densitometry Radiation dose: what you need to know John Damilakis, PhD Associate Professor and Chairman University of Crete, Iraklion, Crete, GREECE Estimation of bone status using X-rays Assessment
Radiation Dosimetry in Digital Breast Tomosynthesis March, 2015 William J. O Connel, Dr. Ph, Senior Medical Physicist Imagination at work. Syllabus 1. Introduction 2. Dosimetry in Mammography 3. Dosimetry
Aquilion ONE: Pediatric Imaging Richard Mather, PhD Senior Manager, CT Clinical Science Toshiba America Medical Systems, Inc. The use of CT in pediatric diagnostic procedures has increased significantly
Low Dose Era in Cardiac CT DIANA E. LITMANOVICH, MD Department of Radiology Beth Israel Deaconess Medical Center Harvard Medical School Disclosures Neither I nor my immediate family members have a financial
Scan for mobile link. Computed Tomography (CT) - Sinuses Computed tomography (CT) of the sinuses uses special x-ray equipment to evaluate the paranasal sinus cavities hollow, air-filled spaces within the